Click to Purchase and Download this Standard or Test Method
Revision Date: July 18, 2013
Updated On: September 17, 2013
4.1 Wet film thickness measurements of coatings applied on articles can be very helpful in controlling the thickness of the final dry coating, although in some specifications the wet film thickness is specified. Most protective and high performance coatings are applied to meet a requirement or specification for dry film thickness for each coat or for the completed coating system, or for both.
4.2 There is a direct relationship between dry film thickness and wet film thickness. The wet film/dry film ratio is determined by the volume of volatiles in the coating as applied, including permitted thinning. With some flat coatings the dry film thickness is higher than that calculated from the wet film thickness. Consequently, the results from the notch gage are not to be used to verify the nonvolatile content of a coating.
4.3 Measurement of wet film thickness at the time of application is most appropriate as it permits correction and adjustment of the film by the applicator at the time of application. Correction of the film after it has dried or chemically cured requires costly extra labor time, may lead to contamination of the film, and may introduce problems of adhesion and integrity of the coating system.
4.4 The procedures using notched gages do not provide as accurate or sensitive measurements of wet film thickness as do the Interchemical and Pfund gages described in Test Methods D1212. Notch gages may, however, be used on nonuniform surfaces, like concrete block, that are too rough to use the Interchemical and Pfund gages. Also notched gages can be very useful in the shop and field for determining the approximate thickness of wet films over commercial articles where size(s) and shape(s) are not suitable for measurements by other types of gages. Examples of such items are ellipses, thin edges, and corners.
4.5 An operator experienced in the use of a notched gage can monitor the coating application well enough to ensure the minimum required film thickness will be obtained.
4.6 Application losses, such as overspray, loss on transfer, and coating residue in application equipment, are a significant unmeasurable part of the coating used on a job and are not accounted for by measurement of wet film thickness.