Click to Purchase and Download this Standard or Test Method
Revision Date: April 26, 2012
Updated On: May 14, 2012
This practice describes three operational steps necessary to ensure accurate coating thickness measurement calibration, verification and adjustment of coating thickness measuring gages, as well as proper methods for obtaining coating thickness measurements on both ferrous and non-ferrous metal substrates.
Many specifications for commercial and industrial coatings projects stipulate a minimum and a maximum dry film thickness for each layer in a coating system. Additionally, most manufacturers of high performance coatings will warranty coating systems based upon, in part, achieving the proper thickness of each layer and the total coating system. Even if a project specification is not provided, the coating manufacturer’s recommendations published on product data sheets can become the governing document(s). Equipment manufacturers produce nondestructive coating thickness testing gages that are used to measure the cumulative or individual thickness of the coating layers, after they are dry. The manufacturers provide information for the adjustment and use of these gages, normally in the form of operating instructions. The user of this equipment must be knowledgeable in the proper operation of these devices, including methods for verifying the accuracy of the equipment prior to, during and after use as well as measurement procedures.
Protect assets and equipment. Keep facilities open and operational. Maintain schedules regardless of weather. Ensure smooth and continuous production. Easy to erect. Safway is the primary distributor of HAKI® products including HAKISPAN and HAKITEC® 750 – in North America.