Click to Purchase and Download this Standard or Test Method
Revision Date: August 11, 2011
Updated On: August 29, 2011
The ability of a paint or coating to resist deterioration of its physical and optical properties caused by exposure to light, heat, and water can be very significant for many applications. This practice is intended to induce property changes associated with end-use conditions, including the effects of sunlight, moisture, and heat. The exposure used in this practice is not intended to simulate the deterioration caused by localized weather phenomena such as atmospheric pollution, biological attack, and saltwater exposure.
Cautions: Variation in results may be expected when different operating conditions are used. Therefore, no reference to the use of this practice shall be made unless accompanied by a report prepared according to Section 10 that describes the specific operating conditions used. Refer to Practice G151 for detailed information on the caveats applicable to use of results obtained according to this practice.
The spectral power distribution of light from fluorescent UV lamps is significantly different from that produced in light and water exposure devices using other light sources. The type and rate of degradation and the performance rankings produced in exposures to fluorescent UV lamps can be much different from those produced by exposures to other types of laboratory light sources.
Interlaboratory comparisons are valid only when all laboratories use the same design of fluorescent UV device, lamp, and exposure conditions.
Reproducibility of test results between laboratories has been shown to be good when the stability of materials is evaluated in terms of performance ranking compared to other materials or to a control. , Therefore, exposure of a similar material of known performance (a control) at the same time as the test materials is strongly recommended. It is recommended that at least three replicates of each material be exposed to allow for statistical evaluation of results.
Test results will depend upon the care that is taken to operate the equipment according to Practice G154. Significant factors include regulation of line voltage, temperature of the room in which the device operates, temperature control, and condition and age of the lamps. All references to exposures in accordance with this practice must include a complete description of the test cycle used.
Our coatings and linings expertise available 24/7, allowing you to explore the best Sherwin-Williams solutions for oil refineries, shale drilling sites, and offshore. Android users click here. Download today!